The stylus is cleaned at a height of 10mm below the upper surface.
For the automatic cleaning of the stylus, we need the following movements to take place in succession:
|2||75||0||10||Max.||The stylus moves along the activation sensor. Therefore the cleaning session will be started.|
|3||0||0||10||Max.||The stylus stands above the cleaning unit.|
|4||0||0||-8||4 mm/s||The stylus moves to a height of 5mm beneath the upper surface. The cleaning of the stylus will be started.|
|5||0||0||-18||4 mm/s||The stylus moves to a height of 15mm beneath the upper surface.|
|6||0||0||50||4 mm/s||The stylus stands above the cleaning unit.|
|7||0||0||50||Max.||End of the cleaning process.|
- The movements [Step 4] [Step 5] [Step 4] are supposed to be repeated during 20 seconds with a movement speed of 4mm/s.
- During the first 10 seconds the stylus will be cleaned with a detergent fluid. The detergent fluid is blended with air by the nebulizer.
- Other movements however may take place with a higher speed.